The relevant parameters of Model 2000 converter have been set according to
User requirements (such as: scale range, density, temperature, pressure and engineering unit, etc), we do not encourage users to preset or modify other parameters, especially the operation of Sensor Zero Trim and Sensor Full-Scale Trim, this menu must be operated by using special instruments and programs under the guidance of engineers, otherwise, it may cause turbulence or other defects.
Actually the only work users have to do, before the flowmeter has been installed, and 24VDC power source has been provided to Model 2000 converter according to two-wire system, and before the flowmeter has been put into use (i.e.: under the condition that there is no flowing liquid in the process pipe),is to have a Sensor Zero Posit-N Adjust, then the instrument can be put into use.
Panel display illustration
When Model 2000 converter is under normal display status, the screen will alternately display the instant flow volume value and accumulated value and the engineering unit (the upper item displays the flow volume value, the lower item displays engineering unit), as shown in Fig 10.
2、Parameter setting and operation flowchart